Operating and service manual for the HPA Synthesizer/Function Generator. Complete with schematics. Hewlett Packard A Synthesizer. HP A Front View. Hewlett Packard A Synthesizer (click images to enlarge). The manual is available from Didier’s. View and Download HP A calibration manual online. Synthesizer/Function Generator. A Synthesizer pdf manual download. Also for: b.
|Published (Last):||4 August 2016|
|PDF File Size:||6.56 Mb|
|ePub File Size:||14.5 Mb|
|Price:||Free* [*Free Regsitration Required]|
Page 8 TB a 50 Hz. TB Test instrument parameters Amplitude accuracy with offset attenuation range 1 into 50 load Attenuator accuracy these errors are additive with the amplitude accuracy errors Amplitude output Amplitude option high voltage output Dc offset Dc offset option Dc plus ac Not verified below 50 Hz. Don’t show me this message again. Page of 23 Go.
Frequency Accuracy 4 Set oscilloscope controls to measure ramp retrace time from the 90 to 10 percent points.
HP 3325A User Manual
By pressing ‘print’ button you will print only current page. Test instrument parameters Harmonic distortion relative to fundamental at full output Amplitude modulation Square wave Ramp retrace Frequency Sine wave Square wave Triangle Positive slope ramp Negative slope ramp Phase offset Amplitude Amplitude accuracy with no attenuation attenuator range 1 into 50 load no dc offset Flatness with no attenuation Ramp retrace time will be 3 s.
Paragraph Page Final procedure TB 4 Enter The manufacturer’s manuals and TM were used as the prime data sources in compiling these instructions. Audio analyzer indication will be dB.
3325A Synthesizer/Function Generator
Record frequency counter indication. Page 22 TB Figure 5. Got it, continue to print.
Calibration Process TB 6. Amplitude Modulation Distortion a. The following peculiar accessory is also required for this calibration: To be distributed in accordance with IDNrequirements for calibration procedure TB Personnel should become familiar with the entire bulletin before beginning the calibration. Square wave amplitude and flatness Items of equipment used in this procedure are referenced within the text by common name as listed in table 2.
No adjustments can be made. If frequency counter does not indicate between Page 12 TB 15 Set frequency counter controls to measure frequency. Page 17 a 1 kHz. If you find any mistakes or if you know of a way to improve the procedure, please let us know.
HP A Service Manual download, schematics, eeprom, repair info for electronics experts
Multimeter will indicate between 2. TB 15 Set frequency counter controls to measure frequency. This manual also for: TB Figure 5. Sine wave amplitude and flatness You can help improve this publication. Add to my manuals Add.
Multimeter will indicate between 4. Enter text from picture: By Order of the Secretary of the Army: The instructions outlined in paragraphs 6 and 7 are preparatory to the calibration process.
Forms, records, and reports TB a 50 Hz.